5 Reasons Your Breast Pump is Losing SuctionFriday, 07 May 2021 07:00 AM
Do you want to guess how many times I have answered this question? So, so many times. Which means you are in good company if you have this same question!
Breast pumps are complicated. They have lots of pieces. They must be put together correctly for the pump to work properly.
Breast pumps are operated by humans. Tired humans. And humans who sometimes feel like they need an extra hand (or two or three) to pump effectively.
In this blog, I am specifically addressing causes of decreased suction with the Evenflo Feeding Advanced Double Electric Breast Pump.
This is where I start when I’m helping a mom, like you, figure out why her pump isn’t sucking:
Plug your pump motor into the outlet and turn it on. Place your finger over the hole that the tubing adapter plugs into. You should feel suction.
Plug your tubing adapter into the pump motor. If you are pumping on one breast, you MUST cap off one side of the adapter or your pump will have no suction. To cap off one side, pull the tubing off of the adapter on the side closest to the cap, then insert the cap.
When you turn the pump on, the diaphragm should squeeze and release.
The valve and membrane should be firmly attached to the bottom of the flange. Without the valve and membrane, your pump will have no suction.
The flange needs to form a complete seal on your breast. While you are learning to use your pump, please remove your nursing pads and take off your bra. It is very common for your bra or nursing pad to sneak between your breast and the flange on the underside of your breast, where you can’t see. This will cause your pump to have no suction. Removing your bra and nursing pads prevents this problem.
Nine times out of ten, the breast pump will suck correctly by addressing one of the above items.
If you’ve gone through this list and the suction still seems off to you, reach out for support. We have a whole team of helpers to get you back on track.